Estimation of temperature distribution with the use of a thermo-camera
Languages of publication
The determination of three-dimensional (3D) temperature distribution within tissue during thermoablation is necessary to estimate procedure efficiency. The use of a thermo-camera combined with finite element modelling is discussed.The temperature distributions in a metal phantom and an animal tissue sample were simulated. In the real experiment, temperatures were measured around the heating probe by a thermo-resistor set and the temperature distributions on samples' surface were acquired by a thermo-camera. The temperatures measured in the experiment were compared with the simulated ones. The differences between the measured and simulated temperatures were lower than 1.3°C and 3.0°C for a metal phantom and tissue sample, respectively.Good agreement was achieved for homogenous material of well-defined parameters. Higher discrepancies for the tissue sample are due to in-homogeneity and to difficulties with describing tissue thermal properties.The proposed method permits the precise prediction of a 3D temperature distribution in in-vitro studies. Potential application for in-vivo procedures requires further investigations.
1 - 1 - 2008
29 - 12 - 2008
- Barauskas R, Gulbinas A, Barauskas G. Investigation of radiofrequency ablation process in liver tissue by finite element modeling and experiment. Medicina. 2007; 43: 310-325.
- Chang I. Finite Element Analysis of Hepatic Radiofreqiency Ablation Probes using Temperature-Dependent Electrical Conductivity. Biomed Eng OnLine. 2003; 2: 12-29.[PubMed][Crossref]
- Dick E, Taylor-Robinson S, Thomas H, Gedroyc W. Ablative therapy for liver tumours. Gut. 2002; 50: 733-739.[Crossref][PubMed]
- Garrean S, Hering J, Helton S, Espat J. A primer on transarterial, chemical, and thermal ablative therapies for hepatic tumors. Am J Surg. 2007; 194: 79-88.[PubMed][WoS][Crossref]
- Haemmerich D, Chachati L, Wright A, Mahvi D, Lee F, Webster J. Hepatic Radiofrequency Ablation With Internally Cooled Probes: Effect of Coolant Temperature on Lesion Size. IEEE Trans Biomed Eng. 2003; 50: 493-500.[PubMed][Crossref]
- Hildebrandt B, Wust P, Ahlers O, Dieing A, Sreenivasa G, Kerner T, Felix R, Riess H. The cellular and molecular basis of hyperthermia. Critical Rev Oncol Hemat. 2002; 43: 33-56.[Crossref]
- Jain M, Wolf P. A Three-Dimensional Finite Element Model of Radiofrequency Ablation with Blood Flow and its Experimental Validation. Ann Biomed Eng. 2000; 28: 1075-1084.[PubMed][Crossref]
- Johnson P, Saidel G. Thermal Model for Fast Simulation During Magnetic Resonance Imaging Guidance of Radio Frequency Ablation. Ann Biomed Eng. 2002; 30: 1152-1161.[PubMed][Crossref]
- Mohammed Y, Verhey F. A finite element method model to simulate laser interstitial thermo therapy in anatomical inhomogeneous regions. Biomed Eng OnLine. 2005; 4: 2-17. doi: 10.1186/1475-925X-4-2.[PubMed][Crossref]
- Reddy J, Gartling D. The Finite Element Method in Heat Transfer and Fluid Dynamics. London: CRC Press; 2003.
- Shang D. Free Convection Film Flows and Heat Transfer. Berlin: Springer; 2006.
- Stauffer P, Goldberg S. Introduction: Thermal ablation therapy. Int J Hyperthermia. 2004; 20: 671-677.[Crossref][PubMed]
- Strasberg S, Linehan D. Radiofrequency Ablation of Liver Tumors. Curr Probl Surg. 2003; 40: 459-498.[Crossref][PubMed]
Publication order reference